Details on affected Devices:
Affected devices: Progressa® Bed Systems P7500
Serial Numbers: S159AW9192 through S305AW2321
Production Dates: June 8, 2017 to November 1, 2017
Description of the Problem: Progressa® Beds manufactured between the dates specified above may experience a problem with the braking system. A component used within the braking system may malfunction, resulting in one or more brake pedals either not engaging or becoming stuck in the “brake” position. If the brake is not engaging, there is a risk for unwanted bed movement which could possibly result in patient or user injury; this failure mode is most prevalent during patient transfers from one surface to another or during patient egresses.